# FTIR spectroscopy

Fourier transform infrared (FTIR) spectroscopy is an advanced and more user-friendly version of infrared spectroscopy. In this article, we have tried to comprehensively summarize the fundamental working principle, the usage, and applications of FTIR spectroscopy. Additionally, we have also discussed how is FTIR spectroscopy different from conventional dispersive infrared spectroscopy. So, let’s start reading to know everything you need to know about FTIR spectroscopy.

## What is FTIR spectroscopy

As its name implies, the Fourier transform infrared spectroscopy is based on a mathematical Fourier transformation operation that breaks down a waveform into its constituent frequencies or wavelengths.

All the characteristic wavelengths of IR radiation (2.5 to 16 μm) are passed through the sample in one go with the help of a Michelson interferometer. Sample molecules absorb a specific wavelength from the entire wavelength range and undergo vibrational energy changes. All the remaining wavelengths are transmitted and recorded as an interferogram. The interferogram is then converted into the conventional IR spectrum via Fourier transformation.

## What is the working principle of FTIR spectroscopy

Below are mentioned all the major components of an FTIR spectrophotometer and how these components work in sync to perform FTIR spectroscopy.

• A blackbody or heated silicon carbide (SiC) is used as the IR radiation source.
• It emits IR radiations in the range of 2.5 to 16  μm (or 4000-625 cm-1).
• These are comparatively low energy (1010 to 1014 Hz) radiations as opposed to UV or visible radiations.

2. Michelson interferometer

• The interferometer consists of a beam splitter (a semi-transparent mirror) along with two reflectors. One reflector is held fixed while the other is movable.
•  As a collimated IR beam falls on the beam splitter, it is split into two beams of equal intensity.
• One beam is transmitted to the fixed mirror while the second is reflected off of the beam splitter to reach the movable mirror.
• Eventually, both the beams are simultaneously reflected by the fixed and movable reflectors such that both coincide at the semi-transparent mirror and constructive interference occurs.
• Different interference patterns are consequently obtained, each denoting a specific wavenumber which is calculated by a highly specialized mathematical software.
• The data obtained is recorded as a detector signal against mirror tilt and it is known as an interferogram.
• A raw signal is first obtained without inserting the sample. Fourier transformation is then applied to convert this interferogram into the reference IR spectrum.
• This process is then repeated with the sample to obtain the required sample spectrum.

3. Sample port

• Latest FTIR spectrophotometers are often fascinated with an attenuated total reflectance ATR port.
• The ATR port makes FTIR spectroscopy more user-friendly. It eliminates the need for complex sample preparation and handling unlike that required for dispersive IR spectroscopy.
• The ATR port is a delicate port made up of a crystal such as diamond, germanium, or zinc selenide (ZnS). The ATR port however must have a higher refractive index than the samples under investigation.
• Samples without any pre-treatment are directly introduced onto the crystal surface.
• IR radiation passes through the crystal and undergoes total internal reflection. This creates an evanescent field in which at a time, only a small portion of the IR radiation penetrates into the sample.
• The intensity of the evanescent field decreases exponentially with time which allows a quick interaction between sample molecules and the IR radiations.
• A specific wavelength is absorbed while the rest are transmitted/ reflected back from the sample to reach the detector.
• The detector uses the transmitted radiation to gain information about IR absorption by the sample. This information is recorded as an electrical signal.
•  An interferogram is obtained which is subsequently transformed into an IR spectrum.

Sampling techniques such as specular and diffuse reflectance can also be used in FTIR spectroscopy, other than ATR. Each sampling technique has its own characteristic strengths and weaknesses.

4. Infrared spectrum

• The IR spectrum is a plot of absorbance against wavenumber.
• Neatly distinguishable sharp peaks obtained in the fundamental group region (4000-1600 cm-1) of the IR spectrum help find the principal functional groups present in the targeted sample.
• Different functional groups absorb characteristic wavelengths of IR radiations. For e.g., a broad peak at 3650 cm-1 on an FTIR spectrum often represents the hydroxyl (OH) functional group. Similarly, the sharp sword-like peaks around 1600-1700 cm-1 are characteristic peaks of carbonyl (CO) functional groups present in aldehydes and ketones.  These peaks appear due to asymmetric stretching vibrations of bonded atoms in the target molecules.

The tongue and sword trick proves quite useful in interpreting an IR spectrum.

• Conversely, the small, heavily entwined peaks in the fingerprint region (1600-625) cm-1 appear due to bending vibrations of targeted molecules by IR absorption.
• The data provided by both fundamental as well as the fingerprint region is finally combined to reveal the identity of the unknown chemical compound.

You can use this standard IR frequency table to identify the different functional groups present in a molecule.

## What information FTIR spectroscopy gives

• FTIR spectroscopy is primarily used for functional group identification and structural elucidation of a variety of chemical compounds.
• FTIR spectroscopy is an important characterization tool in research and development and also in the industrial sector.
• It is used to study newly synthesized polymers, foodstuff, pharmaceutical drugs, petroleum fractions, nanomaterials, biomolecules (polysaccharides, amino acids, proteins ), etc., and for detecting any impurities present in them.
• FTIR spectroscopy can also be performed for studying inorganic chemical substances like minerals, glass, and metal oxides.

Thus, FTIR spectroscopy is a fast, highly efficient, non-destructive, and cost-effective spectroscopic technique that is there to stay for long in the field of chemical structural studies.

If you are interested to know more about spectroscopy, you can also check out our articles:

References

1.Berthomieu, C. and R. Hienerwadel (2009). “Fourier transform infrared (FTIR) spectroscopy.” Photosynth Res 101(2-3): 157-170.

2. Peak, D. (2013). Fourier Transform Infrared Spectroscopic Methods of Soil Analysis. Reference Module in Earth Systems and Environmental Sciences, Elsevier.

### Different Uses and Applications of organic chemistry

The uses and applications of organic chemistry range from life-saving pharmaceutical discoveries to the vibrant colors of fruits and vegetables we see daily. In this

### Ace organic chemistry- tips and tricks, cheat sheets, summary

Mastering organic chemistry takes time and practice. It might seem a daunting task at once if the right approach is not adopted. But in our

### Named reactions of organic chemistry- an overview

This article on organic reactions is a special one in our organic chemistry series. It will guide you through how chemical transformations unlock diversity at

### Organic Spectroscopy

Organic spectroscopy can be used to identify and investigate organic molecules. It deals with the interaction between electromagnetic radiation (EMR) and matter. These waves travel